Abstract：Lumenis holmium laser therapeutic apparatus for medical practitioners has been widely used in clinical treatment due to
its excellent performance. However, in the process of using the holmium laser, there will inevitably be some problems that affect the
clinical use. Maintenance personnel should find out the cause of the fault in time, and take effective methods to repair the fault, then
the holmium laser therapy machine can be restored to normal in the shortest time. Starting from the analysis of the working principle
of holmium laser therapy machine, this paper introduced 3 typical faults maintenance of holmium laser therapy machine in our
hospital, for your reference.
薛莉，赵宏伟，丁军斋，韩晔. 科医人PowerSuite 100W工作原理与典型故障3例[J]. 中国医疗设备, 2020, 35(7): 176-179.
XUE Li, ZHAO Hongwei, DING Junzhai, HAN Ye. Operating Principle and Three Typical Faults of Lumenis PowerSuite 100W. China Medical Devices, 2020, 35(7): 176-179.