Maintenance and Analysis of Fault Code16 for Weinmann
VENTImotion2 Noninvasive Ventilator
SHI Jun1, LIU Hua1, YAO Xufeng2
1. Department of Medical Equipment, Baoshan Hospital of Intergrated Traditional Chinese and Western Medicine,
Shanghai 201999, China; 2. Shanghai University of Medicine & Health Sciences, Shanghai 201318, China
Abstract：Objective To reduce the cost of life cycle management of medical equipment, the fault Code16 of Weinmann
VENTImotion2 noninvasive ventilator was checked and analyzed. Methods Taking the Weinmann VENTImotion2 noninvasive
ventilator as the research object, the Code16 error code was searched without the technical manual, and two cases of maintenance
under Code 16 fault code were analyzed. Results After maintenance, the equipment was tested normally. By self-examinating and
self-repairing, the maintenance cost was reduced, the professional technical level was improved, and the daily maintenance of the
equipment could be promoted according to related problems. Conclusion With the self-repairing of VENTImotion2 noninvasive
ventilator and summarizing of the causes and possibilities of the problems, the maintenance cost of the noninvasive ventilator in our
hospital has reduced and the quality of use has improved.
施骏1，刘华1，姚旭峰2. 万曼VENTImotion2无创呼吸机Code16故障分析与排除[J]. 中国医疗设备, 2020, 35(1): 159-162.
SHI Jun1, LIU Hua1, YAO Xufeng2. Maintenance and Analysis of Fault Code16 for Weinmann
VENTImotion2 Noninvasive Ventilator. China Medical Devices, 2020, 35(1): 159-162.